Forgotten password can be reset at https://www.beat2phone.net/reset-password. Input your username to the text field, and the system will send a password reset code to the email address that is saved in your user profile information. If you don’t remember your username, you can order it to your email from https://www.beat2phone.net/reset-password/RemindEmailForm.
The iOS application does not support professional profile creation in v2.4.4. A professional profile can be restored to the application from the cloud service.
iPhone application does not display the sensor MAC address (visible on the sensor label) in the connecting to sensor dialog. Instead, an Apple proprietary id is shown on the list of sensors to connect to. Make sure that only one sensor is active within the Bluetooth range.
Beat2Phone ECG sensor contains a magnet for controlling the charging connector relays.
Warning: Do not place the Beat2Phone sensor closer than 3cm from a pacemaker. The sensor contains a magnet. The pacemaker may enter into asynchronous mode in the presence of a strong magnetic fields. Normal operation usually continues when the magnetic field is removed.
ICD type of pacemakers may halt operation completely in the presence of strong magnetic fields.
On 19.5.2021, we published a new version of the Beat2Phone ECG mobile application. We also published a new version of the Beat2Phone ECG sensor firmware. The new mobile application version requires the new sensor firmware version to function properly, and without it, there may be problems with recording ECG. We recommend that all users update the mobile application and sensor firmware to the latest version.
Instructions for sensor firmware update:
1. Update the Beat2Phone ECG mobile application. The required version is 2.4.4.
2. Make sure that the sensor battery has enough charge (recommended at least 50%). If the sensor battery is low, the firmware update is not possible.
3. Start the Beat2Phone ECG application in your mobile device. If you are currently recording ECG, stop the recording. Make sure that the sensor is connected to the mobile application. Make sure that the application is in the start view.
4. Take the sensor out of the charger. Put it, for example, on a table, and make sure that your mobile device is close to the sensor.
5. Wait for approximately a minute. The mobile application should notify that a firmware update is available.
6. Start the update by selecting “start update”. The update takes few minutes.
7. Wait until the update is completed.
The mobile application update fixes some bugs related to file deletion and cloud synchronization. The firmware update aims to improve the file transfer process between the sensor and the mobile device. The update makes it possible to also use the sensor with iOS devices, which was not previously possible.
We apologize for the inconvenience.
Measure you chest size below the pecs/breasts. Adjust the Beat2Phone ECG strap to be significantly shorter than you chest size for tight fit to achieve good electrode skin contact.
80cm chest 65-68 cm strap
100cm chest 72-77cm strap
130cm chest 84-93 cm strap
The strap can be adjusted from 65-93cm to fit chest 78-145cm. Instructions for shortening or lengthening the belt are found below.
To shorten the strap, pull a loop (A) of the strap through the adjustment clip (B) so that the lenght (L) is decreased. Lock the loop (A) with a safety pin (C) to prevent it from slipping back through the clip (B).
To lenghten the strap, take a strip (22cm) of cloth ribbon, fold it in half (A) and attach it to the looped end of the strap with a safety pin. You get a 103cm long strap that extends up to 155 cm. In case more extension is needed, take a longer strip of cloth.
If you encounter a problem while using our device or application, it’s best to send an email to [email protected] Our experts will reply to your message as soon as possible. To help identifying and resolving the issue, we recommend that you send us the following information in your message, if possible.
As accurate as possible description of the problem.
Are you using the application on an Android device or an iOS device?
Mobile device model. (for example, Samsung Galaxy S10 or iPhone 10)
Android/iOS operating system version. (This can be found from phone settings, usually from “System” menu.)
Beat2Phone ECG application version. (This is visible in the application home view, e.g. V2.4.3)
Sensor version, Beat2Phone Sensor or Beat2Phone ECG Sensor? (This can be found from a sticker that is located behind the sensor.)
Are you using the cloud service and are you able to log in to the cloud service?
The username that you’re using in the Beat2Phone ECG application. (This can be found from the profile menu that opens from the top right corner of the application home view.)
If you want to quickly remove all the user profiles and recordings from your mobile device, you can delete the Beat2Phone ECG application and then reinstall it. If the profiles reappear when you have uninstalled and reinstalled the Beat2Phone ECG application and this is not wanted, it is because the mobile phone stores your application data to the Google cloud and automatically reloads them when you reinstall the application. To disable this functionality, you must de-select “automatic restore of settings and information” from your phones settings. This is usually found in “backup & restore” menu in the settings.
Unstretched: 65-90 cm
When sufficiently stretched for use: 74-115 cm
The Beat2Phone ECG sensor uses a rechargeable battery that is charged in the included Beat2Phone ECG charging connector.
The battery charges from empty to full in approximately one hour.
In some mobile devices, the device location must be turned on to connect to the sensor.
From the Beat2Phone ECG application settings, check that your sensor is connected to the mobile device. Try disconnecting the sensor by choosing “Forget sensor” and connect to the sensor again. If this does not resolve the issue, try rebooting the mobile device Bluetooth connection and the Beat2Phone ECG application. If still not resolved, try rebooting your mobile device. Finally, reboot the sensor by briefly charging it.
From 24 hours up to 48 hours if the sensor battery is fully charged.
Many mobile devices have, by default, a rather aggressive battery consumption optimization, which can turn off applications that appear inactive. This battery optimization sometimes turns off the Beat2Phone application, which causes the measurement to stop. Usually, this optimization can be adjusted from the mobile device settings, depending on the model of the device. The most reliable measurement can be performed when the mobile device is not used for anything else during the measurement. Even more reliable performance can be achieved if the application remains active, and the mobile device is being charged. To make sure that the measurement has not stopped, it is recommended to check the status of the measurement from time to time.
The Beat2Phone ECG application has not been tested on all mobile devices.
It has been found to work well in the following devices:
In the following devices, Beat2Phone does not work properly (Bluetooth problem):
You can find the short user guide from the instructions for use page >>
You can find instructions for use from the instructions for use page >>